Current Position:Home>>Kits>>ELISA Kits>>QuicKey ELISA>> QuicKey Human RBP4(Retinol Binding Protein 4) ELISA Kit
|Detection range||3.13-100 ng/mL|
|Sample type &Sample volume||serum, plasma, urine; 50μL|
|Specificity||This kit recognizes Human RBP4 in samples. No significant or interference between Human RBP4 and analogues was observed.|
|Reproducibility||Both intra-CV and inter-CV are < 10%.|
|Application||This ELISA kit applies to the in vitro quantitative determination of Human RBP4 concentrations in serum, plasma, urine.Please consult technical support for the applicability if other biological fluids need to be tested.|
|Item||Specifications||Storage conditions after test|
|Micro ELISA Plate (Dismountable)||96T: 8 wells ×12 strips48T: 8 wells ×6 strips||2-8℃, 1 month|
|Reference Standard||96T: 2 vials48T: 1 vial||Discard unused reconstituted standard and dilutions|
|Reference Standard & Sample Diluent||1 vial, 20 mL||2-8℃|
|Biotinylated Detection Ab Working Solution||1 vial, 6 mL|
|HRP Conjugate Diluent||1 vial, 14 mL|
|Concentrated Wash Buffer (25×)||1 vial, 30 mL|
|Concentrated HRP Conjugate (100×)||96T: 1 vial, 120 μL
48T: 1 vial, 60 μL
|2-8℃ (Protect from light)|
|Substrate Reagent||1 vial, 10 mL|
|Stop Solution||1 vial, 10 mL||2-8℃|
|Plate Sealer||5 pieces|
|Product Description||1 copy|
|Certificate of Analysis||1 copy|
Intra-assay Precision (Precision within an assay): 3 samples with low, mid range and high level Human RBP4 were tested 20 times on one plate, respectively.
Inter-assay Precision (Precision between assays): 3 samples with low, mid range and high level Human RBP4 were tested on 3 different plates, 20 replicates in each plate.
|Intra-assay Precision||Inter-assay Precision|
The recovery of Human RBP4 spiked at three different levels in samples throughout the range of the assay was evaluated in various matrices.
|Sample Type||Range (%)||Average Recovery (%)|
Samples were spiked with high concentrations of Human RBP4 and diluted with Reference Standard & Sample Diluent to produce samples with values within the range of the assay.
1. Add 50μL standard or sample to the wells, immediately add 50μL Biotinylated Detection Ab working solution to each well. Incubate for 90 min at 37°C
2. Aspirate and wash the plate for 3 times
3. Add 100μL HRP conjugate working solution. Incubate for 30 min at 37°C. Aspirate and wash the plate for 5 times
4. Add 90μL Substrate Reagent. Incubate for 15 min at 37°C
5. Add 50μL Stop Solution
6. Read the plate at 450nm immediately. Calculation of the results
From now on, if you have published a paper by using any of our products since 1/1/2019, fill out the “Elabscience Publication Reward Application Form”carefully and send it to email@example.com, we will get back to you with the reward after we confirm it ASAP!