Synonyms:NPPA, ANF, ANP, ATFB6, ATRST2, CDD, CDD-ANF, CDP, PND, Atrial natriuretic peptide, natriuretic peptide A
Detection Range: 7.81~500 pg/mL
Sensitivity: 4.69 pg/mL
This ELISA kit applies to the in vitro quantitative determination of Human ANP concentrations in serum, plasma and other biological fluids.
This ELISA kit uses the Competitive-ELISA principle. The micro ELISA plate provided in this kit has been pre-coated with Human ANP. During the reaction, Human ANP in the sample or standard competes with a fixed amount of Human ANP on the solid phase supporter for sites on the Biotinylated Detection Ab specific to Human ANP. Excess conjugate and unbound sample or standard are washed from the plate, and Avidin conjugated to Horseradish Peroxidase (HRP) are added to each microplate well and incubated. Then a TMB substrate solution is added to each well. The enzyme-substrate reaction is terminated by the addition of stop solution and the color change is measured spectrophotometrically at a wavelength of 450 nm ± 2 nm. The concentration of Human ANP in the samples is then determined by comparing the OD of the samples to the standard curve.
|Detection Range||7.81—500 pg/mL|
|Sample Type||Serum, plasma and other biological fluids|
This kit recognizes Human ANP in samples. No significant cross-reactivity or interference between Human ANP and analogues was observed.
As the OD values of the standard curve may vary according to the conditions of the actual assay performance (e.g. operator, pipetting technique, washing technique or temperature effects), the operator should establish a standard curve for each test. Typical standard curve and data is provided below for reference only.
Intra-assay Precision (Precision within an assay): 3 samples with low, mid range and high level Human ANP were tested 20 times on one plate, respectively.
Inter-assay Precision (Precision between assays): 3 samples with low, mid range and high level Human ANP were tested on 3 different plates, 20 replicates in each plate.
|Intra-assay Precision||Inter-assay Precision|
|C V (%)||6.34||5.23||4.52||5.86||5.73||5.51|
The recovery of Human ANP spiked at three different levels in samples throughout the range of the assay was evaluated in various matrices.
|Sample Type||Range (%)||Average Recovery (%)|
|EDTA plasma (n=5)||94-104||99|
|Cell culture media (n=5)||92-107||99|
Samples were spiked with high concentrations of Human ANP and diluted with Reference Standard & Sample Diluent to produce samples with values within the range of the assay.
|Serum (n=5)||EDTA plasma (n=5)||Cell culture media (n=5)|
An unopened kit can be stored at 4℃ for 1 month. If the kit is not used within 1 month, store the items separately according to the following conditions once the kit is received.
|Micro ELISA Plate(Dismountable)||8 wells ×12 strips||-20℃, 6 months|
|Reference Standard||2 vials|
|Concentrated Biotinylated Detection Ab (100×)||1 vial, 120 μL|
|Concentrated HRP Conjugate (100×)||1 vial, 120 μL||-20℃(shading light), 6 months|
|Reference Standard & Sample Diluent||1 vial, 20 mL||4℃, 6 months|
|Biotinylated Detection Ab Diluent||1 vial, 14 mL|
|HRP Conjugate Diluent||1 vial, 14 mL|
|Concentrated Wash Buffer (25×)||1 vial, 30 mL|
|Substrate Reagent||1 vial, 10 mL||4℃(shading light)|
|Stop Solution||1 vial, 10 mL||4℃|
|Plate Sealer||5 pieces|
|Product Description||1 copy|
|Certificate of Analysis||1 copy|
Note: All reagent bottle caps must be tightened to prevent evaporation and microbial pollution.
The volume of reagents in partial shipments is a little more than the volume marked on the label, please use accurate measuring equipment instead of directly pouring into the vial(s).
1.Add 50 μL standard or sample to each well.
2.Immediately add 50 μL Biotinylated Detection Ab to each well.
3.Incubate for 45 min at 37℃. Aspirate and wash 3 times.
4.Add 100 μL HRP Conjugate to each well. Incubate for 30 min at 37℃. Aspirate and wash 5 times.
5.Add 90 μL Substrate Reagent. Incubate for 15 min at 37℃.
6.Add 50 μL Stop Solution.
7.Read at 450 nm immediately. Calculation of results.
M**********nSubmitted [ Jun 15 2018 ]
Asked: Does this kit cross-react with mouse ANP?
adminSubmitted [ Jun 15 2018 ]
Answered: Hi Marina, Thanks for your inquiry. The cross-reactivity with the mouse ANP is 22.4% in our lab as the identity of the immunogen sequences is high. Any other questions, pls feel free to contact us. Best regards.